This calibration manual focuses on performance verification and calibration of the Fluke. A and A Digital Multimeters (hereafter referred to as the. The Fluke A & A digit precision multimeters have the accuracy & versatility to handle your most demanding measurements, on the bench or in a. Manual/User Manual on CD-ROM, FVF-BASIC. FlukeView Forms Software Basic Version. Ordering Information. Fluke A. Digit Precision. Multimeter.
|Published (Last):||22 September 2009|
|PDF File Size:||13.67 Mb|
|ePub File Size:||10.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
To replace the Current Input fuses, 1. First verify that the line voltage selection is set to the correct position, and then that the correct fuse for that line voltage is installed. Page 2 Fluke authorized resellers shall extend this warranty on new and unused products to end-user customers only but have no authority to extend a greater or different warranty on behalf of Fluke.
Accessories Table lists the available accessories for the A and A. To make a capacitance measurement: Basic Electronic Test Mannual Set.
Using the Analyze Functions The Meter is capable of performing mathematical operations with measured values, as well as tracking a series of measurements. Front-Panel Operation Introduction Introduction The Meter can be controlled either by sending commands through one of its communication interfaces or by manually operating its front panel controls. Appendices Errors Incorrect packet size from inguard Info packet rec’d; link not active Inguard Calibration Constant write failed Inguard not responding recv Inguard not responding send INITiate received but was ignored Instrument configuration load failed Instrument configuration store failed Insufficient memory Invalid dimensions in a channel list Invalid parameter Invalid parameter Of course, these DMMs are also durable and dependable, features you expect from any Fluke meter.
If you want to store the present Meter configuration in a location other than the first four locations, press the MORE soft key. For the A, the lower four keys select different functions; see inset. If zero is not used, add 0. Making a Two-wire Resistance Measurement To make a two-wire resistance measurement: Press the 1 A or 0.
Page 9 Page Don’t show me this message again. Setting The Display Resolution Front-Panel Operation Configuring the Meter for a Measurement The beeper state is stored in non-volatile memory and does not change when the Meter is turned off or after a remote interface reset.
Introduction and Specifications Safety Information Table Create customized, professional looking documents that can be printed or emailed. Table of Contents Add to my manuals Add. Resistance Introduction and Specifications Electrical Specifications Resistance Specifications are for 4-wire resistance function, 2 x 4-wire resistance, or 2-wire resistance with zero.
Fuse, mA, V, Fast, 0.
8845A/8846A 6.5 Digit Precision Multimeters
To enter the M value: To start a histogram session: If the number of samples is set to a value greater thanconsecutive memory files will be used to store all the samples. The display indicates the next available memory location. To store the present Meter configuration in one of the first four memory locations, press the appropriate soft key. Press the MORE soft key.
Connect the Meter to a properly grounded three-prong outlet. manuual
Replacement Tips for TL Multiple connectors give you maximum flexibility Choose from several interfaces to connect the A or A to a personal computer: The copy process will take several seconds. Register Log in Contact Us.
Digit Multimeter | DMM | Fluke A & A
The Fluke A and A 6. The manual is divided into the following chapters: Histogram Display In addition to the bar graph, the left third of the display also displays the number of samples, the average, and standard 8846s values. If the Meter appears damaged or something is missing, contact both the carrier and Fluke immediately. An example of the diode test display is shown below. Displays the filter menu.
Fluke 8845A User Manual
Front-Panel Operation Controlling Trigger Functions Note Setting the Meter to trigger via a remote command is only accessible through the remote interface. One contact fkuke connected to HI or LO input circuits and the other contact is connected to the Sense input circuits. After removing the power-up configuration, power the Meter down and back up using the rear-panel power switch to 8846 factory default settings.
Understanding Specifications for Precision Multimeters Press B to go back to the previous menu. Connecting To Line Manjal First verify that the line voltage selection is set to the correct position, and then that the correct fuse for that line voltage is installed.
An example of the capacitance display is shown below. DC Voltage, Clock Resistance, setting date and time, Firmware Configurations checking revision, recalling, Fluke storing, contacting of, Configure for Measurements, Frequency meausrements, Continuity Front Panel, Setting of threshold, Front Panel Navigation, Input Lo and ma or 10A connectors.
Selects between manual and auto range mode. Opening the cover of the Meter to make this alteration may void the Crest factor0. To check a diode: USB port on the front panel of the Meter.